Summary of Features
The basic MIRADAS concept is a near-infrared multi-object echelle spectrograph oper- ating at spectral resolution R=20,000 over the 1-2.5µm bandpass. MIRADAS selects targets from a 5 arcmin field of regard using up to 12 deployable probe arms with pick- off mirror optics, each feeding a 3.7 x 1.2 arcsec field of view to the spectrograph. The spectrograph input optics also include a "slit slicer" which reformats each probe field into 3 end-to-end slices of a fixed 3.7 x 0.4-arcsec format – combining the advantages of minimal slit losses in any seeing conditions better than 1.2-arcsec, while at the same time providing some (limited) two-dimensional spatial resolution. The spectrograph optics then provide a range of configurations providing the observer with the ability to choose between maximal multiplex advantage and maximal wavelength coverage, with several intermediate options, depending upon the needs of the science program.
|Target field of regard||5-arcmin diameter||Each probe arm patrols a 2D workspace within this circular field|
|Individual target field of view||3.7x1.2-arcsec|
|Slit slicer geometry||3 slices of 3.7x4.0-arcsec each|
|Detector focal plane||4096x2048 pixels||Mosaic of 2Kx2K HAWAII-2RG|
|Continuum sensitivity||J=18.0 mag
|S/N=10 for 1-hour on-source exposure|
|Emission line sensitivity||5x10-18 ergs/cm2/s
8x10-18 ergs/cm2/s (resolved)
|S/N=10 for 1-hour on-source exposure; resolved source assumes 1 square arcsecond detect cell|
|Configuration||MXS Targets||Instantaneous Bandpass|
|Medium MXS J||3||1.17-1.34 µm|
|Medium MXS H||3||1.50-1.82 µm|
|Medium MXS K||3||1.92-2.49 µm|
|Maximum MXS||12||Any SINGLE order from the Table below|
MIRADAS Echelle Orders:
|Order||Wavelength (µm)||Band||Order||Wavelength (µm)||Band|
Note that additional combinations of multiplex and wavelength coverage are also possible. By selecting different order!sorting filters and probe output masks, the observer could choose any combination of target multiplex (NMXS) and wavelength coverage by consecutive orders (Norder), as long as the product NMXS * Norder < 12. For instance, a configuration with 6 MXS targets and a wavelength coverage from 2.0885 to 2.3820 microns (orders 15 and 16) would work.
Subject: Kick-off for the MIRADAS instrument
Recently a contract was signed between GRANTECAN and the University of Florida on the design of a new instrument for the GTC. The instrument, MIRADAS, will be a multi-object near-IR spectrograph reaching a spectral resolution of 20000. It is conceived as a common-user instrument that will be located in the folded-Cassegrain focal station.
MIRADAS will be one of the third-generation instruments for the telescope and is led by Prof. Stephen Eikenberry from the University of Florida (USA), in collaboration with Universidad de Barcelona, Universidad Complutense de Madrid (UCM), Instituto de Astrofísica de Canarias (IAC), Institut de Física d'Altes Energies (IFAE), Institut d'Estudis Espacials de Catalunya (IEEC), and Instituto de Astronomía de la Universidad Nacional Autónoma de México (UNAM). The science team behind the instrument comprises a large number of scientists from 8 institutions within the GTC community.
The capability of MIRADAS to work at infrared wavelengths where obscuration by gas and dust between the stars is less of a problem, its multiplexing capability of observing several objects at the same time, and the high spectral resolution that can be reached make the instrument extremely attractive to tackle a wide range of scientific questions. All this, combined with the massive light-gathering power of the GTC provides us with a recipe to achieve very exciting scientific goals that will expand our knowledge in several key areas of astrophysics.
Examples of such key areas are the detailed study of massive stars in our galaxy that will help us understand galactic structure and the chemical history of our Milky Way. MIRADAS will also have an important impact for galaxies in our cosmic neighborhood, such as for instance in the study of massive black holes that lurk at the centers of many of these galaxies, and of the dynamical behavior of galaxies. The instrument also contemplates polarimetric measurements which open up a unique window in areas such as the study of cirmumstellar disks, planet formation, and even the possibility of mapping the surface structures of cool stars.
If all goes well for this very demanding instrument design, the science community could star to take profit from the capability of MIRADAS by 2017.
MIRADAS contact persons at GTC
|contact||email @ gtc.iac.es|
|Stefan Geier - main contact||stefan.geier|
Last modified: 02 December 2016